|CoPrime® system overview
||Benefits of the CoPrime® system
|Option 1: 5 inlet manifold on 1 pump for multiple buffer processing||Option 8: Post-column pH conductivity
A conductivity sensor (with temperature compensation) and a pH sensor are fitted in the same flow cell, which is integrated in line between the column and outlet ports. Two ranges of conductivity are defined, one for the process and another for cleaning.
|Option 2: Second pump with 2 inlets enabling:
||Option 9: Post-column UV dual wavelength
With this option, a mono-wavelength UV absorption sensor, Optek® AF45 sensor, or a dual-wavelength with an Optek® AF46 sensor is proposed.
|Option 3: Mass flowmeter
One mass flowmeter is set up between the pump and the bubble trap for monitoring and regulation of the flow fed to the column.
Note: This option includes only one flowmeter. If option 2 (second pump) is selected, the flowmeter option is to be selected twice. The mass flowmeter option excludes the mag flowmeter option.
|Option 10: Five outlets
Three additional 3/4 in. outlet ports with NovAseptic® valves are added downstream of the column to separate different fractions.
|Option 4: Valve block of pre-column filter either disposable or in a housing
Note: Filter and housing not included.
|Option 11: CIP manifold for cleaning the entire system (excluding the column) in CIP, from one single inlet to one single outlet|
|Option 5: Valve block and pre-column filter housing
Note: Series 3000 housing 20 in. I/O TC 3/4 in.
— Type 226 Code 7, Recessed Base
|Option 12: Feedback switches on NovAseptic® valve
With this option, every valve is mounted with an integrated position feedback switch. Closed position is physically detected.
|Option 6: Pre-column pH and conductivity
A conductivity sensor (with temperature compensation) and a pH sensor are fitted in the same flow cell, which is integrated in line between the filter and the column.
|Option 13: Electromagnetic flowmeter
The performance of the flowmeter is dependent on the conductivity of the product. Product conductivity must be greater than 5 μS/cm for liquids, in general, and greater than 10 μS/cm for demineralized water.
|Option 7: HETP inlet block valve
This option enables injection of the HETP sample as close as possible to the column inlet, through a NovAseptic® divert valve.
Note: It is possible to perform an HETP sequence using an inlet port instead of the dedicated divert valve.
|Option 14: Filter hoods
The addition of stainless steel filter hoods on the enclosure filters and fan allows an IP rating of IPx5.